Compared to other low flux imaging technologies, an Pushing Sensitivity to the Brink: Selecting the Right Imaging Technology for Your Application published in Biophotonics.shows high quantum efficiency ( ) and little , much like traditional . More, it presents negligible readout noise akin the intensified ( ), thus providing the best sensitivity in poor lighting conditions. For more information about the various low light imaging cameras available today, refer to our article
However, like all imaging technologies, thesuffers from thermal and readout noises as well as other electronic-dependant factors that affect its sensitivity. Clock-induced charges ( ), the prime noise source in , particularly at high readout rates, may contaminate images up to 200 times more than the when only a handful of photons reaches the camera sensor.
Listed below are the primary parameters that, when unaddressed, lessen the sensitivity of low light imagingsystems, along with the means by which Nüvü Camēras inventively and elegantly overcomes such obstacles.
Ideally, ansensor would only detect . However, the mere thermal agitation within the chip is sufficient to eject electrons from the silicon body, which are then collected by the potential wells. Even in total darkness, the wells slowly gather these charges, also referred to as dark electrons, and do so in an increasing amount with prolonged exposure times. It is as if the camera had been exposed to a faint light source.
Thedetector is cooled down to temperatures nearing -90 °C to counter such noise, known as either or . Doing so substantially decreases the number of dark electrons within the chip, hence improving dramatically the signal. Performing acquisitions in the inverted mode of operation ( ) further reduces the contribution. Refer to the Modes of Operation section later in this tutorial for more information.
Nüvü™ addresseswith either liquid nitrogen or a thermoelectric cooling (TEC) unit to absorb the detector’s heat. To that end, Nüvü Camēras has developed a revolutionary camera packaging that can maintain the detector’s temperature at -85 °C with an extraordinary precision of 0.01 °C even when performing fast acquisitions and reading the entire detector frame. Such stability ensures the uniformity of the signal amplification through the electron-multiplying register, as the number of dark electrons remains the same See the subsection for more information.
Readout noise occurs whenever the charges are amplified and digitized. Negligible in high illumination conditions, as in the order of 2 to 10 electrons, the readout noise becomes comparable to thesignal in near-darkness scenes.
technology includes a charge amplification stage, the electron-multiplying (EM) register, which boosts the signal before digitization. As this stage increases the incoming signal by a factor up to a few thousand, the readout noise becomes negligible.
The factor associated with theamplification is known as the . Boosting the signal, however, comes at a price: the EM registry is a sensitive component that may easily saturate, and saturation may lead to its premature aging or damage.
Note that the EM registry boosts not only thesignal but also dark electrons and clock-induced charges (see the next subsection). Nüvü™ addresses such issue with both the innovative Controller for Counting Photons ( ) and unrivaled cooling performances. The first minimizes the clock-induced charge contribution while the latter cuts down levels. Therefore, Nüvü™ is the sole camera manufacturer that offers cameras supporting up to 5000.
For most applications, Nüvü™ recommends employing EM gains below 1,000 to avoid dynamic range losses. The camera’s dynamic range, defined as the ratio between the full-well electron capacity and the camera overall noise level, declines with increasing EM gains as a result of the charge amplification register capacity. For increased gains, the EM register accepts fewer electrons before saturation due to the amplification of the charges contained in an entire row and not individual pixel charges. But lower gains lead to better dynamic ranges, which refine the image contrast.
Nevertheless, for applications requiring—further on this page—Nüvü Camēras strongly recommends values of more than 3,000 to optimize the sensitivity and dynamic range.
Shifting theat high readout rates requires the application of both high voltages and high-frequency clocks, the latter signal coordinating the displacement within the sensor. With greater clock frequencies are associated a supplementary noise source known as the clock-induced charges ( ).
Fortunately, Nüvü Camēras innovative and patentedController for Counting Photons ( ) generates finer and more adjustable clock signals than those produced by other controllers. As such, it drastically limits the creation of while reading the at high speeds.
Somemay be left behind during the charge transfer process, especially at high readout speeds: the electron shifting process from one potential well to the next may be incomplete, thus creating so-called deferred charges in the imaging region. Consequently, the leftover electrons artificially increase the brightness of certain pixels and decrease the overall image quality of the camera.
Nonetheless, on top of lowering, the Controller for Counting Photons enhances charge transfer efficiency ( ), and does so at any operating temperature and readout frequencies, hence leading to images of highly superior quality.
The avalanche process that boosts thesignal before readout is unpredictable in nature: one can only ascertain the mean value of the , never its exact value. The signal follows a Poisson probability distribution, as seen in the figure below.
Uncertainties arise when attempting to determine the exact number of electrons that contributed to the signal before amplification: as shown above, the electron distributions overlap considerably despite the number of electrons n, thus making it impossible to ascertain how many electrons entered the EM register first. Such ambiguity results in the contamination of-acquired images by the excess noise factor ( ), which has a √2 value at high . This stochastic noise source, noticeable in low light conditions, affects an image’s signal-to-noise ratio ( ) as if the sensor’s quantum efficiency was cut in half.
Performingmeasurements eliminates the by assessing the number of per pixel and attributing to each a value of 1 or 0. The charge amplification registry, therefore, cannot change the pixels’ output and thus not alter their values with the electron-multiplication associated uncertainty. Further explanations are provided in the Photon Counting section.