The Electron-Multiplying( ) technology renders readout noise negligible thanks to , which allows reaching significantly higher frame rates than conventional by using higher pixel rates. Other than the readout rate, the number of pixels read as well as the size of the sensor will affect the frame rate.
Moreover, thesensor’s architecture also gives it a high level of flexibility to create custom readout sequences. This allows to further maximize frame rate when using only portions of the sensors (Regions of Interest – ROIs) or even to reach extremely low exposure times, down to the few microseconds, by minimizing the movement of the charges on the sensor.
The structure of anacquisition is highly important in understanding what limits frame rate. use a “frame transfer” architecture, which makes use of 2 regions on the sensor: an imaging area and a storage area. While the imaging area is exposed to light, the storage area is shielded, which enables the readout of the of a previous exposure even while acquiring a new image. Consequently, the shortest exposure possible for the imaging area is limited by the processing time of the image, when the storage area is free.
Once the imaging area has been illuminated for the selected exposure time, the first step after exposure is to rapidly transfer the charges from the exposed area to the storage area. The larger the sensor the more rows on the sensor, the longer this frame transfer takes (~10 us to 1 ms) as there are more rows of charges that must be transferred. The second step is to read all the pixels. The readout time will depend both on the number of pixels to read and the sensor’s readout rate (~0.5 ms to 100 ms for a full sensor). As such, two elements affect frame rate: the size of the sensor (number of pixels to read & move on the sensor) and the readout rate (rate at which the pixels are read).
In most acquisition modes, the readout time will be the main factor limiting the maximum frame rate. Higher readout rates can be used to speed up the process, butreadout is a complex process and care must be taken to preserve image quality at higher speeds. Thanks to better control over the readout & noise sources, Nüvü Camēras is the only manufacturer to offer the highest readout rate of 30 MHz (30 million pixels per second) on all sensor formats.
Another way to reduce readout time is to reduce the number of pixels to read. This is achieved by using a Region of Interest () which restricts readout to section of the detector relevant to the acquisition. Nüvü™ also supports the use of multiple regions of interest ( ) for more flexibility. Using pixel binning is another way to increase frame rate, although it is not recommended in low-light acquisitions since the resulting super-pixel includes the noise of all sub-pixels whereas there is not necessarily signal in each sub-pixel.
An important particularity of ROIs and binning is that the horizontal dimension (i.e. the number of lines) of theor super-pixel has no influence on imaging speeds. This means a 32 x 32 or 32 x 512 and 2 x 2 binning or 2 x 1 binning achieve the same frame rate. This is because when a line is selected to be read, the entirety of line must be shifted pixel-by-pixel through the horizontal multiplication register even if they are not of interest.